Thank you so much for replying ! First reference - failed to open - sometimes Omega site do not wish to be browsed by Chrome... Second reference refers to Win versions of cards .... I have to use DOS.
At the meantime, have upgraded DOS 5.0 to 6.22 version. Also got answer from Omega dept in EU, and there is reference
where on the page is another reference to omega ( seems, same ftp ) server with OMBPERD488 DOS driver and installation manual ( OMG - 488 pages !!! ):
I tried to install this diver . Temproraly have removed GPIB card from PC to check all switch configuration and then used this info to configure GPIB driver:
- IRQ5 interrupt;
- 02E1 I/O address;
- DMA channel - 3.
Still no communication, here is POST screen. What is strange for me. Device ( masspectrometer ) software have not asked any questions regarding the I/O during the installation. If I am right, It means that software has some unknown for me fixed I/O parameters. Have no idea how to get them.....